- Cerumen Management
- Clinical Tools & Supplies
- Diagnostic Equipment & Supplies
- Fitting Supplies
- Hearing Conservation
- Impression Materials
- Infection Control
- Office & Educational Items
- Otoscopes & Headlamps
- Resale Accessories
- Assistive Devices
- OAKTREE TV
- OUTLET STORE
Amplified Stethoscopes: Hearing Instrument Programming Considerations
A.U. Bankaitis, PhD
Medical professionals rely on auscultation to routinely examine the status of the circulatory, respiratory, and/or gastrointestinal systems. Auscultation is defined as listening to internal sounds of the body and represents an essential component in the delivery of healthcare services. The procedure is accomplished through the use of a stethoscope, a medical device specifically designed to enable physicians, nurses, and other medical professionals to detect and analyze heart, lung, and/or bowel sounds for purposes of differential diagnosis. For medical professionals with hearing loss, performing auscultation while wearing hearing instruments requires a collaborative effort between the medical practitioner and the audiologist to ultimately secure a successful outcome. The purpose of this white paper is to outline hearing instrument programming considerations as it relates to auscultation.
Frequency and Intensity Characteristics of Auscultation:
While there appears to be some discrepancy regarding the exact frequencies of various internal body sounds, breath/lung sounds may range from 70 Hz to 4000 Hz (Cardionics, 2008) with most falling below 2000 Hz (Noland, 2009). The most critical breath/lung sounds used for differential diagnosis fall in the 200 Hz to 600 Hz frequency range (Abrams, 1987). In contrast, heart sounds fall in the 20 Hz to 650 Hz frequency range (Rennert, Morris, & Barrere, 2004), although the most essential heart sounds for differential diagnosis fall in the 70 Hz to 120 Hz frequency range (Noland, 2009; Cardionics, 2008).
In the absence of intensity data of body sounds, attempts were made to quantify breath/lung sounds from CD recordings provided by an amplified stethoscope manufacturer. Due to limitations associated with access to calibration data of recordings and associated specifications related to microphone and instrumentation methods, the most accurate information available is the general statement that breath/lung and/or heart sounds are very soft sounds.
Hearing Instrument Programming Considerations for Medical Professionals Motivated to Use Hearing Instruments with an Amplified Stethoscope:
Often times the listening needs of the practitioner as dictated by audiometric configuration will significantly differ from the listening needs of that same practitioner for routine auscultation procedures. As visually portrayed in Figure 1, while the various audiometric configurations dictate a high frequency emphasis program for optimizing speech, audiologists will need to be prepared to provide a low frequency emphasis program for medical professionals with hearing loss since both breath/lung and heart sounds are low frequency sounds. Specifically, as indicated by the different shaded areas, lung sounds may range up through about 4000 Hz although the most important breath/lung will fall in the blue shaded area; similarly, the red shaded area represents the most important frequency range for heart sounds.
Figure 1: Audiometric configurations representing high-frequency hearing loss with shaded areas reflecting critical frequencies for performing auscultation
Furthermore, the intensity of both breath/lung and heart sounds are extremely soft. Ideally, a hearing instrument utilizing Wide Dynamic Range Compression would be beneficial. Experimenting with compression ratios in the low frequencies whereby adjustments are made to yield a 2:1 or 3:1 compression ratio despite the presence of normal low frequency hearing may be beneficial.
- Create a separate auscultation program for the medical professional
- The frequency response of the auscultation program must maintain a low frequency emphasis, regardless of audiometric hearing needs; this will require setting low frequency settings as low as possible (100-250 Hz) since critical breath/lung and heart sounds range from 200-600 Hz and 70-120 Hz respectively
- Recognize that the intensity of body sounds are very soft; even in the presence of normal low-frequency hearing, adjusting WDRC ratios to 2:1 or 3:1 may be helpful
Counseling a medical professional on the use of an amplified stethoscope with their hearing instruments during auscultation procedures remains similar to the counseling new hearing instrument wearers receive. Depending on the type, degree and configuration of the hearing loss, breath and/or heart sounds may not/will not sound like something they have been accustomed to hearing. For example, body sounds may actually sound more like clicking noises and making the medical professional aware of this may help them recognize and re-learn what they need to hear. Furthermore, identifying the most optimal amplified stethoscope solution for the medical professional with hearing loss requires access to and the ability for the patient to try two or three options.
Abrams, J. (1987). Essentials of cardiac physical diagnosis. Philadelphia: Lea & Febiger.
Cardionics (2008, April). Use of Phonak Savia hearing aids with E-scope electronic stethoscope. (Technical Information Bulletin issued by Cardionics, www.cardionics.com).
Noland, M. (2009, September). Stethoscopes and hearing aids (Technical white paper issued by Cardionics), www.cardionics.com.
Rennert, N., Morris, F. & Barrere, C. (2004, February). How to cope with scopes: stethoscope selection and use with hearing aids and CIs. Hearing Review, p. 34-38.